|Format||4K x 4K 15-micron|
|FOV (arcmin)||26.5 x 26.5|
|Pixel size (1x1 binning, arcsec)||0.389|
|Readout modes||full frame, 1 x 1 binning|
|central 2k x 2k window, 2x2 binning|
|Cycle time (readout+overhead; 1x1 binning, s)||28.0|
|Pixel full well (e-)||100,000 to 120,000 typical, unbinned, see note on binned readout below.|
|Read noise (e-)||7.5 to 8 e- typical|
|Gain (e-/ADU)||1.0 after pipeline processing|
|Dark current (e-/pix/s)||0.002 @ -100 C (estimated)|
Note: The gain and saturation limit are reflected in the FITS header keywords GAIN and and SATURATE, respectively, and those are set for each individual camera in the network. The units of the saturation limit will be ADU for raw images, and electrons for BANZAI-processed images.
Description of Standard filters
The CCDs attach below a Filter/Shutter unit with carbon fiber disk shutter and 3 wheels, each 8 positions, providing
An exposure time calculator can be used to estimate typical exposure times for our different aperture classes.
There are 4 off-axis ports which can all see light simultaneously with the main science camera.
One port to the North of the main science field always has an FLI autoguider with 0.34" pixels and 5.7' Field of View. We intend to deploy on least one telescope per site a Lucky Imaging/High Speed Photometer (LIHSP) consisting of:
LCO has introduced a new readout mode for the Sinistro cameras on the 1-meter telescopes. The mode reads the central 2048 x 2048 pixels, with 2x2 pixel binning, and with a readout overhead of 9 seconds.
Time-sequence observations of bright objects become less efficient with longer CCD readout times and other overheads. An example of a science program that benefits from reduced readout overheads is the observation of planet transits. Also, phenomena that change on time scales on the order of minutes cannot be observed efficiently if the readout overhead is too large. A short readout time will open a window into new science.
LCO has deployed a new CCD readout mode at the 1-meter telescopes where only the central 2k x 2k pixels of Sinistro images are read out, and the pixels are binned 2 x 2, resulting in a 1k x 1k pixel image. To shorten the per-image overhead further, the number of CCD wipes before each exposure are reduced compared to full-frame readouts. The "central 2x2k binned" mode reduces the total overhead from ~28 s per image to ~ 9 s.
The shortened overhead can increase the shutter open time over the duration of an observation, and hence benefits the achievable signal-to-noise. This mode is recommended for the acquisition of time series where:
The new central_2k_2x2 mode is released in shared risk mode. Observers should validate if this new mode is appropriate for their science before committing all allocated time.
In the portal's request composition page, the readout mode is selected from a drop down menu in the Instrument Configuration section.
In the API, include the following statement in the new API language to explicitly select a readout mode:
"mode": "full_frame" || “central_2k_2x2”}]
If no readout mode is defined, the full_frame mode will be used per default.